ASHRAE110-1995 (ANSI approved) is one of the most challenging standards in the world that tests the containment efficiency of a fume hood. The efficiency is derived from the fume hood's ability to contain the released challenge gas under normal operation. The ASHRAE110-1995 test facility at Esco is constructed based on recommendations given by Invent-UK, a third party organisation that certifies fume hoods. The ASHRAE 110-1995 is mainly composed of three tests: flow visualization, face velocity uniformity and SF6 containment.
Face Velocity Test
Tracer Gas Containment
The derived performance as determined by the smoke visualization test and the face velocity uniformity measurement provides a valuable indicator of the containment property of the fume hood. The SF6 containment test provides the ultimate proof in establishing this sequence. If the results of the smoke test and the face velocity test are good, then the containment capability of a fume hood is good.
Here in Esco, we are capable of conducting the complete ASHRAE110-1995 test in our laboratory. For further information on the test method and observed results, please download detailed report.
View video (US)
View video (Asia)
View video (Europe)
EN14175 is comprised of a series of containment tests other than the normal face velocity tests. Containment tests include Inner Measurement Plane Tests, Outer Measurement Plane Tests and Robustness of Containment Test. The challenge gas which is released into the fume hood is 10% SF6 and 90% N2. A set of sampling probes is positioned at various predetermined locations to monitor SF6 escaping from the hood.
For the inner and outer measurement plane tests, a detector probe set is moved across the sash opening at various heights and sash positions to assess the overall containment efficiency of the fume hood under normal operations. For the robustness test, a vertical plank mounted perpendicular to the sash plane is made to run to and fro across the sash opening at a fixed speed. This disturbs the inflow and gives an opportunity for the SF6 to escape. A reliable fume hood ensures that at that level of disturbance, the fume hood is still able to quarantine the released SF6.
For conformity evaluation, a type test is carried out on one of Esco's EFH units fresh off the production line. Thanks to Esco's design, EFH passed the performance requirements of EN14175.
Esco Micro (M) Sdn. Bhd.
No. 15, Jalan Sungai Buloh 27/101A, Seksyen 27, 40400 Shah Alam Selangor, Malaysia
+603-5634 5833 | +603-56387895
13-1, Jalan Melaka Raya 32, Taman Melaka Raya 1, 75000 Melaka
606-284 0007 | 606-283 0288